A Liquid-Freon Permeameter
- B.G. Hurd
- Document ID
- Society of Petroleum Engineers
- Society of Petroleum Engineers Journal
- Publication Date
- March 1962
- Document Type
- Journal Paper
- 18 - 20
- 1962. Original copyright American Institute of Mining, Metallurgical, and Petroleum Engineers, Inc. Copyright has expired.
- 5.5.2 Core Analysis, 5.2.1 Phase Behavior and PVT Measurements, 4.1.5 Processing Equipment, 1.10 Drilling Equipment, 5.6.2 Core Analysis, 4.1.2 Separation and Treating
- 5 in the last 30 days
- 211 since 2007
- Show more detail
- View rights & permissions
HURD, B.G., SOCONY MOBIL OIL CO., INC., DALLAS, TEX.
A liquid-Freon permeameter suitable for making routine permeability determinations on small plug samples is described. The instrument is characterized by simplicity of design and ease of operation, and can be assembled inexpensively from stock items of laboratory equipment. It combines the unique advantages of both liquid and gas permeameters while eliminating many undesirable features of both general classes of instruments. Precision and accuracy of the specific Freon permeability measurements compare favorably with results of conventional liquid and gas permeameters.
The liquid-Freon permeameter described in this paper was designed and built for special investigations of the properties of petroleum reservoir rocks. However, its unique advantages over conventional gas and liquid permeameters make it eminently suitable for routine use in the core analysis of small plug samples. In general, gas permeameters are very popular because of their simplicity of design and ease of operation. However, apparent gas permeabilities are normally higher than true specific permeabilities unless special corrections for gas-slippage are applied. Since these corrections require two or more measurements at different internal gas pressures, much of the time advantage of the specific gas permeability measurements is lost. Liquid permeability measurements, on the other hand, require no slip correction, but liquid saturation of the sample is time-consuming and difficult to insure. Often, too, the saturating liquid must be extracted or dried from the plug before it can be used for another experiment. The liquid Freon permeameter described herein retains the inherent accuracy of the liquid permeameter, while affording the speed and ease of operation of the gas permeameter. Freon-12 (dichlorodifluoromethane) is a halogenated hydrocarbon, immiscible in the liquid phase with water. It may be considered as an oil phase in laboratory fluid-flow studies. Like butane and propane, it is a gas at normal room temperatures and atmospheric pressure but can be liquefied at moderate pressures. Complete liquid saturation of a plug sample with any of these fluids can be easily achieved by condensing vapors in the sample and operating the permeameter at a pressure above the vapor pressure of the liquid. Freon-12 is superior to butane or propane for permeability measurements only in that it is relatively non toxic and completely incombustible, and thus presents no fire or explosion hazard. Freon-12 is believed to be completely nonreactive with most petroleum-reservoir rock samples. It volatilizes immediately from a plug on depressurizing to leave a clean, dry sample suitable for subsequent experiments. Thus, the inherent risk of changing physical properties of rock samples by cleaning and drying operations between permeability measurement and subsequent experiments is eliminated.
DESCRIPTION OF THE PERMEAMETER
The design of the liquid Freon permeameter is shown schematically in Fig. 1.
|File Size||207 KB||Number of Pages||3|